Message 6 of 6. Everything else was left as default. Known Issues This driver does not have any known defects. You are running into Dot-Comma-issues. I get error or 0xBFFE saying that the session handle is not valid.
|Date Added:||10 December 2009|
|File Size:||63.67 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
Install The installation package for the driver is distributed as an MSI 2.
The logical name is hence a “modifiable DSSP selector”. You can download the latest version from: Message 2 of 8. Message labvjew of 9. Message 3 of 8.
This operation actually runs: The RS port is incorrectly configured. Any insight or tips for resolving these issues would be appreciated as I keep searching for them. The program starts to run and takes an initial voltage measurement, but once the agildnt gets to setting the output voltage to the device the program shows the s3644a mentioned.
Or does it sound like it is an issue with the device instead of the VI?
However, when using Labview 7. This doesnt seem to be a program issue, but rather a communication between Labview and the Agilent power supply.
Unable to get hpexa Labview dirvers to work – Discussion Forums – National Instruments
Everything else was left as default. Known Issues This driver does not have any known defects. I’m not sure what you mean by a “better method for getting data over an RS connection”.
It seems a communication error. You can install the driver by double-clicking on the file. What baud rate are you running at?
How to set up logical names for IVI class drivers – Discussion Forums – National Instruments
There are no public properties which can be set from the command line. Message 3 of 6. Message 1 of 9. NET, Service Pack 2 is required. These three strings are values of properties in the IIviIdentity interface. Most Active Software Boards: This driver supports instruments manufactured by Agilent Technologies. If you install the.
NET Framework itself is not required by this driver. If you post your VI a better analysis can be done. It is also available agient Most Active Software Boards: Message 7 of 9. I believe that you need a null-modem cable as well but verify that in the manual.
Agilent E36xx IVI-COM Driver
If you plan to use the driver with. That does not make any sense. BTW, the above vi can be found at: